The device is intended for the use of pain associated with sore or aching muscles of the lower back, arms, legs, shoulders, or feet due to strain from exercise, work or normal household activities. When used for the symptomatic relief and management of chronic, intractable pain and relief of pain associated with arthritis, use the Tap, Shoulder, Arm or Leg mode of stimulation. Environments of Use: Clinics, hospital and home environments. Patient Population: Adult.
TENS is a therapy that has been used for over 30 years by medical professionals, such as physical therapists and chiropractors. Our TENS devices have various modes which use different methods of moving the electrical currents throughout the muscle in order to administer the treatment. Microcurrent is a therapy that applies extremely small (less than one microampere) electrical currents to the nerves using electrodes placed on the skin.
These devices can be used separately. Additionally, Device A can administer a different treatment type and length than Device B.
If doing two therapies, the pads should be a minimum of one inch apart from each other. The pads should never touch or overlap.
It is recommended that one session using either the TENS or Microcurrent should last 30 minutes. Treatment times up to 60 minutes for TENS and 180 minutes for Microcurrent are an option given within the device as well. The maximum amount of TENS treatments you should have in one day is three, while you can do an unlimited number of sessions in a day of Microcurrent.
Connect the AC adapter plug to the AC adapter jack on the charger. Then simply insert the AC adapter into an outlet. Finally, set the TENS device on the charger to initiate charging.
The rechargeable battery will last up to 500 uses when fully charged under the following conditions: the new battery was fully charged and used one time per day in normal temperatures. Overall, the battery life will depend on usage and storage conditions. Please Note: If the device hasn’t been used for more than three months, charge the device.
In the Therapy Setting Screen on the app, where you select which part of the body you want to treat, the devices have their own respective A and B icons at the top of the screen. To the right of the A and B icons, a tiny battery symbol can be seen. Here is where you can see if your device is fully charged, half full, or low. Please also refer to the Instruction Manual under “Parts Names and Functions of the Omron TENS App”. There you will see a chart showing all of the symbols for these various levels of battery and what they will look like on the app screen.
The device that is connected first will be referred to as device “A” on the app. The device that is connected second will be referred to as device “B”. When you select either device A or B on the application, the corresponding device will blink Green.
For first time users, remove the pad from the sealed plastic bag. To attach the device to the pads, attach the connector to the connection hole and push the device until it clicks into place. You can also refer to the Instruction Manual.
With proper storage and use, you may extend the life of your pad to 30 uses. The life of the pads may vary due to skin and storage conditions. New pads can be purchased by calling 866.216.1333 or online at OmronHealthcare.com.
To clean the ADHESIVE side of the pads, wipe the surface with a soft cloth moistened with water. Be sure not to use a paper substitute, as this will leave behind more debris. To clean the NON-ADHESIVE side of the pads, wipe the surface with a soft cloth moistened with water or a neutral detergent, then wipe the surface with a dry, soft cloth to remove extra water.
First, turn the device off. Clean the device with a lightly moistened cloth and wipe gently. DO NOT use chemicals (like thinner or benzene). DO NOT let water get into the internal area of the device.
You can refer to page 12 the Instruction Manual for a chart listing the meaning of each Display of Indicator Lights. Click here for a quick link to the Instruction Manual.
In the Therapy Setting Screen where you select which body location you want to treat, select the area of the body you want to treat with the respective device. Then, tap the icon on the screen that looks look like a pad used for the device with a question mark inside of it. By tapping this icon, you will be able to see the proper pad placement for the respective area of the body in which you selected to treat.
During treatment on the online application, you will see a “+” sign along with a “-” sign. During the therapy session, click either the “+” sign to increase intensity or “-” to decrease the intensity level.
Yes, press the icon located in the top right of the treatment (right under Device B) that looks like a phone being rotated from Portrait to Landscape view. Once clicked, you will see both therapies at once in a Landscape view.
Tap the red outline of the heart, while using a specific therapy. Once you have saved it to your Favorites, the heart will appear fully red. The button for selecting the therapy as a Favorite can be seen in the top left hand corner of the app during use.
You can view your past treatments in your Treatment/Session Log. It is located in the Settings tab located on the bottom of the app next to the Notepad symbol. Once you click the Settings tab, you will be able to see a Treatment/Session Log tab located at the bottom. Click that, and it will show your previous treatments and pain history recorded for each treatment.
Yes, the device power will automatically turn off 10 minutes after the therapy is complete, the pad has been removed from your body, or the therapy has been paused.
To reset the device you must press the Reset button located on the back of the device with a pen or something with a tip the size of a safety pin.
This symbol is displayed when there is a notification in the app. It will appear in the right corner above the device name (i.e. A or B) and/or in the right corner above the help button, which appears at the bottom of the screen to the right of the gear or Settings button and is symbolized by a question mark.
You can view Help Information within the app by tapping the question mark icon that appears at the bottom of your screen to the right of the gear or Settings icon.
Dispose of the device and battery in accordance with the local government regulations.